The Incu-Shaker™ Mini is the most compact shaking incubator in its class. Despite the remarkably small footprint, a large 11.5 x 9.5 in. work space accepts several platforms and a variety of flasks, tubes and other common vessels.
The standard platform (included) features a non-slip, rubber coated surface, ideal for tissue culture flasks, Petri dishes and staining trays.
The convenient MAGic Clamp™ platform is also available for use with Erlenmeyer flasks and test tube racks. The unique, magnetic attachment method is the easiest way to instantly change between flask clamps of different sizes.
Other platforms are also available. For details on all Incu-Shaker™ Mini platforms, click here.
|Platform Area (ft^2)||0.76|
|Platform Dimensions||9.5" x 11.5"|
|Max Load (lb)|
|RPM - Max||300|
|RPM - Min||30|
|Orbit Diameter (mm)||19 mm|
|Weight (lb)||25 lb|
Speed: Variable, 30 to 300 rpm
Orbit: 19mm / 3/4 in.
Temp. Range: Ambient +5 to 70°C
Temp. Increment: +0.1°C
Temp. Accuracy: +/- 0.1°C
Temp. Uniformity: +/- 0.25°C
Speed Increment: 1 rpm
Platform Dimensions: 9.5 x 11.5 in./ 28 x 41 x 33 cm
Operating Temp. Range: +4 to +65°C
Dimensions: 11 x 18.8 x 13 in. / 28 x 48x 33 cm (w x d x h)
Weight: 25 lbs (11.5kg)
Height, Lid Open: 25 in. / 63 cm
Timer: 1 min. to 48 hours / continuous
Electrical: 115 or 230V, 50-60Hz, 300W
Warranty: 2 Years
Is the MAGic Clamp™ platform included with purchase of the Incu-Shaker™ Mini?
No, all MAGic Clamp accessories are sold separately.
What is the speed accuracy of the Incu-Shaker Mini?
The accuracy is +/- 5% of the set speed.
What is the resolution of the speed settings?
The speed is adjustable in 1 rpm increments.
Is the Incu-Shaker™ Mini UL listed?
No, it is not UL listed.
Is the Incu-Shaker™ Mini suitable for continuous operation?
Yes, it is. It has a continuous operation setting and can be run 24/7 without issue.
Does the Incu-Shaker™ Mini allow the use of rotation or heat without the other? In other words, can you only use the heat without shaking or only use shaking without heat?
Yes, you can use either shaking or heat without the other by setting the parameter you do not want to use to zero.3059 Benchmark Scientific 2018-08-22